1 Max 1200°C design , continuous operating at 1100°C .
2 High quality alloy resistance wire as heating element
3 High quality Alumina Ceramic Fiber as chamber material
4 Temperature controller: PID temperature control
5 Chamber capacity : 1 Liters to 45 Liters according to customer
6 Thermcouple : K style
7 Application : Laboratories of institutions of higher
learning.Scientific research institutions factory enterprise. New
material field ( metal material, ceramic material, nano
material,semiconductor material etc.)
1000℃,1200℃Vacuum Atmosphere Box Furnace
STZ series products mainly aim at daily application and development
of the laboratory.High quality chamber material and stable
temperature control system can ensure the reliability of
experimental data.New type ceramic fiber material as the chamber
material and high quality high temperature alloy wire as heating
element.Temperature controller adopts large touch screen
microcomputer PID control module,which can reach accurate
temperature control and constant temperature requirements.
1.Chamber material adopts vacuum forming ceramic fiber.It has not
powder drops off with high temperature,small heating capacity and
saving energy more than 50%.
2.Heating element adopts high temperature alloy wire,which can bear
heavy load.It is stable and has long service life
3.High heating speed,and it takes 15-30 minutes from room
temperature to 1000℃
4.High precision of temperature control with the temperature
compensation and correction function at precision±1℃
5.Imported temperature control instrument with the programmable
function can set heating curve and programme 2 groups 16 segments.
6.Integrative structure can reduce the used space with excellent
7.Electronic parts and components adopt DELIXI brand with earth
leakage function.It is safe and reliable.
8.With alerting signal for the over-temperature during the working
process,which can complete protective action automatically.
9.When the programs are set and press the starting button,the
following steps will proceed automatically.
10.Double door design structure with flange sealing and PTFE
sealing ring can insure sealing property and can inlet inert
protective gas under vacuum.The external metal pipe can make the
high temperature volatile hazardous gases flow to the specified
11.Optional large screen paperless recorder is available which can
complete the real time record of the heating curve.It can analysis
and print experimental data with memory card.
(Φ x Length)
|Heating length||Voltage||Power||Heating element|